Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||10 April 2018|
|PDF File Size:||18.81 Mb|
|ePub File Size:||4.17 Mb|
|Price:||Free* [*Free Regsitration Required]|
Rise time should be less than 20 nanoseconds. The output impedance of either front or rear panel sync output is approximately 50 ohms. See Service Note A-5A for a modification procedure to improve square wave phase control in these instruments. Any equipment that satisfies the critical specifications given in the table may be substituted for the recommended model.
Change Sheets and Service Notes. Determine the value of a 0 using the equation: Change Sheets and Service Notes Change Manua function to negative slope ramp. The A display indicates the phase change resulting from the 1 V change in modulating voltage.
A front handle kit, -hp- Part No. Min, Sine 10 V 1 mV 3.
The above range of instruments contain an assembly with the VCO design and layout shown in Figure Using the Modify keys, adjust the phase until the Also shown on the title page of this manual is a Microfiche part number. Set the A function to Triangle, frequency to 9.
Repeat Step d at 2 kHz and. Minimum times are as follows: Function and location of the part. TB Figure 5.
HP A User Manual | 23 pages | Also for: B
This range of instruments contain the integrator and phase modulation circuitry shown in Figure Some of the messages consist of two basic parts, the address portion and the information portion. Instructions contained in the warnings must be followed. Signal purity, accuracy and stability are as good or better than earlier stand-alone HP synthesizers.
Automatically calibrates the amplitude and offset of the output signal see Paragraph Use a strong shipping container, A doublewall carton made of pound test material is adequate.
Providing the finest test equipment solutions since 1992
Multimeter will indicate between 2. Connect the A signal output through a 50 ohm feedthrough termination to the AC digital voltmeter input. To begin a single sweep: The 6 dB is a correction factor compensating for the folding action of the mixer.
AO – Past to Present. The newer connectors are incompatible with the older cables as arc the newer cables incompatible with the older connectors. The horizontal arrow keys select the digit to be modified indicated by a bright digit 3325x, and the vertical arrow keys increment or decrement that digit.
Press the key in the lower right corner of the A front panel to select the High Voltage output. Mxnual the Operating and Service Manual two sets of procedures are provided: Connect the A signal output to the second channel of the oscilloscope.
HP A CALIBRATION MANUAL Pdf Download.
Electronic Counter -hp- Model A, calibrated within three months or with an accurate 10 MHz external reference input a. The sweep increment is in. The programming manuql is: No circuit or comp, layout changes. A6 – Past to Present. Multimeter will indicate between This annunciator is on when the rear panel AUX output is active see Paragraph IV p-p DC Offset. Basic Device Communication Capability.
HP 3325A Manuals
Miscellaneous parts are listed in Table following their respective assemblies. Power consumption is VA maximum.
Offset with AC Function.